|●About measurement location
||Home, washing Hoshiba, sandbox, garden, park, athletic ground
||・There are small parts. Please can never enter the entrance. I might suffocate.
・Please do not dismantle this machine.
・Please treat it carefully not to drop this plane.
・Please do not soak it in water.
・When water appears, please wipe off water immediately.
・When you do not use it for a long term, please take off a battery.
||Present location indication range 0.10μSv/h - 19.99μSv/h (in microsievert /)
Quantity of multiplication indication range 0.1mSv - 99.9mSv (millimeters sievert)
Use battery alkali size AAA battery
Battery life continuation 30 days (in the case of use of LED indication one hour a day)
※By a battery to use, life may be mixed up.
External form dimensions 55(W) *80 (H) *20 (D) mm
110 g of mass (at the time of battery wearing)
Less than use temperature range 0 ゜ C - 45 ゜ C 90% RH (case without the dew condensation)
When supplement, a current radiation dose is less than 0.10μSv/h, the LED indication is displayed with ".Lo".
(in ".Lo" a meaning of "it is low")
・When current radiation doses exceed 19.99μSv/h, the LED indication is displayed with ".Hi".
(in ".Hi" a meaning of "it is high")
|Precaution to measure definitely
||As "はかるっち" detects a gamma (γ) line emitted by radioactive material, I cannot measure a right dose of radioactivity when there is an electromagnetic wave having this gamma beam and near property. I recommend that you use it from an electric appliance (TV, microwave oven, wirelessless telephone, radio machine, fluorescent lamp, remote control, AC adapter) producing a powerful electromagnetic wave ten minutes apart. (when a value rises suddenly in the room, the influence of the electromagnetic wave is thought about.)
In addition, please treat it easily as vibration and the shock may be detrimental.
I put "はかるっち" to the place that I want to measure or push the button once with having easily by hand, and, to (a change decreases) that a displayed dose of radioactivity level is stable, just please wait for a while. (with the stability of the dose of radioactivity level, the most loose cash of the indication assumes degree to delicately change an aim.)
When (0.6-5μSv/h) is relatively small degree for 1-3 minutes when (more than 5μSv/h) is relatively big for around one minute when a dose of radioactivity level is very big, for an aim of the measurement time, degree is suitable for (0.5μSv/h or less) for 4-6 minutes.
As "はかるっち" continues for three minutes when I push the button once and displays it, a dose of radioactivity level is very big, or a value is stable while I display it when it is relatively big.
When a dose of radioactivity level is relatively small, you wait until indication goes out once (three minutes) and you push the button once again calmly and measure it for around six minutes in total, and please wait for the stability of the dose of radioactivity level.
As I let "はかるっち" calculates from the radiation dose that you detected and display the latest value every 5 seconds, the measurement (calculation) means things when indication changes sequentially. I will display (the change of the indication level decreases) which a dose of radioactivity level is stable in its and a dose of radioactivity level of the place.
As "はかるっち" always measures it when I do not display it, indication changes frequently to measure a dose of radioactivity level of the new point when I move a place, and to calculate it.
When I measure a dose of radioactivity that is a whit, a number fluctuates by wind.
I recommend that I measure several times and take the mean.
|| A model number: FM-H1W white
A model number: FM-H1P pink
A model number: FM-H1B blue
||ipt'a (イプタ) Support center
Telephone 024-934-4360 () except time in from Monday to Friday from 10:00 to 17:00 (holiday
|●Advertisement responsibility for the contents of an article
||Kochi-ya energy 0242-85-7380